BluetoothBloodPressureDB Appfor Android and iOS supports blood pressure measuring devices for data transmission when these use the general Bluetooth GATT standard for blood pressure measuring devices.
Attention: Not every blood pressure device with Bluetooth supports this standard. We successfully tested the following models (June 2019):
- Beurer BM 54
- Beurer BC 57
- Beurer BM 57
- Beurer BM 77
- Beurer BC 85
- Beurer BM 85
- Medisana BW 300 connect
- Medisana BU 530 connect
- Medisana BU 546 connect
- Medisana BU 550 connect
- Medisana BU 575 connect
Medisana BU A52 connect
- Medisana SL 100 connect
- Omron Evolv (App Version 6.1.27)
- Omron RS7 Intelli IT
- Sanitas SBC 53
- Sanitas SBM 37
- Sanitas SBM 67
- Soehnle systo monitor connect 300
- Soehnle systo monitor connect 400
- Silvercrest SBM 68
Not compatible are for example: devices from ADE, Omron, Withings/Nokia and also the Beurer BM 95 as well as the Medisana MTX connect.
On top of that, your mobile device must support Bluetooth Smart in both the hardware and the operating system (Android from 4.3, iOS since version 5).
In the app settings, you can enable and disable the Bluetooth feature. Is it active, you can see a Bluetooth symbol on the top right. The app automatically responds to incoming connections. It is not necessary to select the new entry.
As soon as a Bluetooth device is detected, the symbol changes color and connects to the device. After establishing a connection, it is then checked whether the device is compatible to send blood pressure data. If so, the data is interrogated while the screen is locked in the app. The data is synchronized directly after transmission. When the small icon changes color in the upper corner, but no data is received, a different Bluetooth smart device, which is sending data is usually close by.
The measuring devices are transmitting after each measurement or by a special function described in the manual. All data stored in the selected user memory of the device is transferred. With the devices from Beurer and Sanitas, all data from the two user memories are always transferred. The app displays the incoming connection and counts the received data records. This automatically detects whether one new measurement is being transmitted or several. The time from the measuring device applies. Unfortunately, the transfer of all data always takes some time. Beurer sends much faster than Medisana.
Omron Evolv: The device has to be paired with the mobile device before the first transfer. For date and time setup the Omron app has to be used.
Omron RS7 Intelli IT: The device always sends the data when the Bluetooth function is activated. Therefore, first perform the measurement and then activate the Bluetooth function in the app settings. The device has to be paired using the Omron app before the first transfer.
Storage space and users:
If data from different user memory cells is received on the device, the app asks which memory cell should be read. The app remembers this setting.
If the transfer of the data from a memory cell deviating from the previous user memory cell is transferred, but not from the previously used memory cell, the app indicates this. This is to prevent the mixing of data from different users.
The app manages these settings with the respective user since the app can manage several user accounts. However, these settings are not synchronized with the server.
If you want to change the storage location on the device and reassign it in the app, please click on additional information on the last transmission. This allows the settings to be reset.
Note: Internally, the counting of the memory cells starts from 0. So the first memory cell has the number 0. Individual devices may differ from that, like for example the BC57. The app then displays the device memory location 1 as 2 and the memory location 2 as 3. This doesn't influence the correct functioning.
Frequent errors and their causes:Data is being transferred, but not present: Please check the date of the measuring device. Especially at the beginning of a year, many inquiries about missing data are being made, which then turn out to be a wrong year set in the measuring device.
Data transfer doesn’t work: Does the transfer work with the manufacturer's app? If problems occur, but your device meets all requirements (operating system/hardware), please try restarting the device.
Another device is interrupting the connection: If the Bluetooth icon is blue, but no data is transmitted, another Bluetooth device is transmitting. The app connects to the device and then detects that it is not a blood pressure monitor. The device ID is stored internally and no connection is established with this device. Nevertheless, in very rare cases it may happen that another Bluetooth device permanently prevents the connection with the blood pressure monitor.
On Android 6, the location recognition (GPS) must be activated to receive data via Bluetooth.
Mail: email@example.com | Contact information | German: BlutdruckDaten.de | BloodPressureDB App